FDA is aware of three recent adverse event reports involving the use of the Vet Jet™ device in kittens. In two of the cases, the user failed or may have failed to properly lock the nozzle into the device before using it. The nozzle subsequently dislodged at the time of vaccination, causing a fracture of the femur (the thigh bone).
Merial is fully investigating the three cases. The company is also revising the “Instructions for Use” for the Vet Jet™ device to stress that users need to securely lock the nozzle into the device. In addition, Merial is sending a letter to all Vet Jet™ device users urging them to carefully follow the nozzle locking procedure.
FDA asks veterinarians and pet owners who want to report problems with the Vet Jet™ device to first contact Merial. They can also report problems to FDA by submitting FORM FDA 1932a.